The humidity transmitters as all other measuring instruments shall be periodically checked and eventually adjusted. The most simple solution for this purpose is the E+E Humidity Calibration Kit.
The E+E Humidity Calibration Kit is a cost effective solution for calibrating humidity measuring instruments with sensing probes Ø 10-12 mm (0.4-0.47 inch). It is very easy to use and it does not require highly qualified technical staff. The kit consists of a humidity calibration chamber and a choice of E+E Humidity Standard Sets.
The E+E Humidity Standards are non-saturated salt solutions available in sets of five or fifty single-use ampoules, which may be stored an indefinite time. The salt solutions are non-harmful, handling them does not require specific safety measures. Safety data sheet is available upon request. Each E+E Humidity Standard Set is supplied with a traceable calibration certificate, issued by the Austrian National Metrology Institute (NMI).
0% RH ... ±0.3% RHhumidity value in %RH ... accuracy at 23 °C (73.4 °F)
5% RH ... ±0.5% RH
10% RH ... ±0.5% RH
20% RH ... ±0.5% RH
35% RH ... ±0.5% RH
50% RH ... ±0.9% RHhumidity value in %RH ... accuracy at 23 °C (73.4 °F)
65% RH ... ±0.9% RH
80% RH ... ±1.2% RH
95% RH ... ±1.2% RH
Send us your e-mail, we will do our best to help you!
Humidity Calibration-Kit Datasheet(428.89 kb)
Calibration Kit - User Guide(1042.14 kb)
What is the difference between a 2 wire and a 3 wire current output?
Transmitter with "two-wires" technology receive the power from the process, and the signal is carry on return wire (closed current loop). With technology "three-wires" the power supply is separate from current output: 2 wires are the power supply and third carry the signal.