The robust, accurate and long term stable EE381 moisture in oil sensor is ideal for in-line monitoring of various oils as well as for predictive maintenance of plant and machinery.
The EE381 measures water activity [aw] and temperature [T], and calculates water content [x] in ppm in transformer, lubrication and hydraulic oil, as well as in diesel fuel.
The measured data is available on two analogue outputs and on the optional LCD display.
The robust metal EE381 enclosure can be rotated for optimal positioning of the display and the status LEDs.
An optional adapter and the free EE-PCS Product Configuration Software facilitate easy configuration and adjustment of the EE381.
- Measurands: water activity aw, temperature T and water content x in ppm
- Oil temperature up to 120 °C (248 °F)
- Two analogue outputs
- LCD display
- Status LEDs
- Pressure rating 20 bar
- G ½" ISO or ½" NPT process connection
- Swirling front-end
- Configurable and adjustable
0...1 awMeasuring range
-40...120 °C (-40...248 °F)
±0.2 °C (±0.36 °F)
0-1/5/10V or 0/4-20mAOutputs
Send us your e-mail, we will do our best to help you!
EE381 Datasheet(902.17 kb)
Scaling of the outputs(252.78 kb)
EE-PCA Product Configuration Adapter(790.77 kb)
EE381 Manual(2675.31 kb)
Cleaning Instructions(1054.21 kb)
Relabeling in case of product configuration change(1339.66 kb)
EE-PCS Product Configuration Software(29086.25 kb)
Oil monitoring for wind turbines
The warranty period of standard gear units for wind energy is usually 2 to 3 years. After the expiry of the warranty period, maintenance teams are responsible for the smooth running of the turbine for the remainder of its service life - which can be up to 20 years
Stainless steel filter
RS232 Interface Cable with Pin Connector
What is the calibration for a specific oil? What are specific oil parameter?
Calibration for a specific oil is a laboratory procedure for determining the saturation curve of the oil. The saturation curved is described by two parameters (A and B). These parameters are used for setting up the moisture in oil sensor for correct calculation of the water content (ppm) out of the measured temperature and water activity (aw).
How can I replace the sensing element of the E+E MIO sensor?
Sensor replacement is possible the EE360 models when option PC4 (pluggable probe). Please see the operation manual for detail and spare part code ordering guide.
How can I clean the sensing element of the E+E MIO sensor?
Please see the cleaning instructions.
Can the coating be used for E+E MIO sensors?
No, coating is not available for MIO sensors.
What kind of calibration certificate is included in the standard scope of supply of the E+E MIO sensors?
The standard scope of supply of E+E MIO sensorsincludes an inspection certificate according DIN EN 10204 - 3.1. For details on calibration certificates please see our white paper "Calibration and Traceability in Measuring Technology".
Are there different A, B parameters for mineral transformer oil from various manufacturers?
No, mineral transformer oils from different manufacturers are quite similar.. A and B parameters are the same.
What is the difference between a 2 wire and a 3 wire current output?
Transmitter with "two-wires" technology receive the power from the process, and the signal is carry on return wire (closed current loop). With technology "three-wires" the power supply is separate from current output: 2 wires are the power supply and third carry the signal.