EE576安装方便快捷，探头杆上的对准线与安装法兰能清楚的指示传感探头安装的方向。安装法兰可随意调节变送器探头的插入深度。探杆内的电子电路能提供一个线形的模拟输出信号：0-5V DC或0-10V DC对应的风速范围可选 0...2m/s 或者 0...1m/s。
0.2... 1m/s: ±(0.05m/s+测量值的2%)
0,2... 2m/s: ±(0.08m/s+测量值的4%)
10-19V DC或19-29V DC供电
Due to its design and materials, the E+E thin film sensing element for air velocity is highly insensitive to pollution. When operaten in very dusty or diry environment, cleaning might be necessary. See section maintenance at downloads.epluse.com/fileadmin/data/miscellaneous/datasheet_air_velocity_measurement_basics.pdf
This is not possible. Return the device to E+E for service.
The standard scope of supply of E+E air velocity sensors includes an inspection certificate according DIN EN 10204 - 2.2 or DIN EN 10204 - 3.1, depending on the model. Please see the product data sheet and operation manual for the type of certificate. For details on calibration certificates please see page 5 in www.kalibrierdienst.at/fileadmin/content/White_Paper_Calibration_and_traceability_in_measuring_technology.pdf
Accredited calibration certificate for air velocity is available upon request, please see our calibration website.
Transmitter with "two-wires" technology receive the power from the process, and the signal is carry on return wire (closed current loop). With technology "three-wires" the power supply is separate from current output: 2 wires are the power supply and third carry the signal.