根据Humor 20的双压法工作原理，它在普通的实验室环境中就能正常工作，而无需昂贵的全空调环境。仪器操作非常简便，工作时只需要蒸馏水、10bar压力的无油压缩空气和90-230V AC的电源。设备本身提供校准仪表所需的24V DC直流电源，与变送器端子的接线快捷方便。
- 10...95% RH
- 0.3 - 0.9% RH
- < 3 min/测量点
- 90...230V AC
下载 Humor 20
The Humor 20 scope of supply includes an inspection certificate according to DIN EN 10204 – 3.1 in the standard scope of supply. An accredited calibration certificate with specific test points is available upon request.
Please mention that you require an accredited calibration certificate when inquiring for a new HUMOR 20 or when returning for service or re-calibration an existing Humor 20. The most common accredited certificate festure 3 calibration points (15/55/90%RH) or 5 calibration points (15/35/55/75/90%RH). Other certificates upon request.
During normal operation mode, incease the humidity generated by the Humor 10 to the maximum by rotating the blue knob counter-clockwise. The HUMOR 20 display will show arround 100% RH, flashing. Open the cover of the distilled water tank and remove the cover from the calibration chamber. If the display shows is 100 %RH ±0.8%RH, the Humor 20 is within specifications. Is the reading is beyond 100 %RH ±0.8%RH please see the operation manual, chapter 6.4.
This is the sensing head from the plausibility sensor from the HUMOR 20.
By design, the temperature inside the measuring and the saturation cambers of the Humor 20 is uniform. By this, in the range 10 to 40°C the influence of the ambient temperature is negligible.
With distilled water inside during transportation, the Humor 20 may be tilted max. 20°. If this cannot be assured, the water must be removed before transportation.
The humidity generator requires a compressed airof max. 10 bar (145psi), filtered from particles > 0.1µm , with residual oil <0.01mg/m³
The humidity generator requires distilled water. Do not ever fill in tap water, because it causes depositsin the piping and pressure regulators, which lead to malfunktion and eventually to the failure of Humor 20.
When you plan to use the humidity generator again within 4 days we recommend to keep the it powered and the humidity level set to 100% rh. For a longer periods out og use, please remove the water and dry the humidity generator before switch of the power.
Set the display switch to temperature
A blinking display after rotationg the blue knob for setting a new RH level indicates that the new RH evel has not been reached yet. The display should stop blinking after about 3 minutes when reached the new RH point. For a longer (or continuously) blinking display please consult the HUMOR20 operation manual.
Yes. Please contact E+E for details and availability.