E+E Elektronik의 EE354는 특히 냉장 드라이어의 노점 모니터링용으로 개발된 제품입니다. -3...20 °C Td (26.6...68 °F Td)범위에서 ±1 °C (±1.8 °F)의 높은 정확도를 유지하며 장기간 안정성이 우수하여 냉장 드라이어 제조사에 적합한 OEM용 모델입니다.
사이즈가 매우 작고 간결하여 좁고 제한된 공간에도 쉽게 설치할 수 있습니다. 스테인리스스틸 소재의 견고한 하우징, ½" ISO 또는 NPT 피팅, 그리고 M12x1 커넥터로 결선 및 설치가 빠르고 간편합니다.
Our calibration service packages ensure that you are on the safe side when it comes to the accuracy and vailability of your measuring devices. Choose the right service package for your E+E product and benefit rom professional device service, calculable costs and excellent measuring accuracy in the long-term. Here you can find more detailed information.
-20...50 °C Td측정범위
± 1 °C Td정확도
Modbus RTU and 4-20mA출력
EE354 Datasheet(1332.58 kb)
Scaling of the outputs(252.78 kb)
EE354 Manual(666.64 kb)
Cleaning Instructions(891.89 kb)
Modbus Application Note AN0103(181.41 kb)
EE-PCS Product Configuration Software(21982.92 kb)
E+E Calibration Service Packages(1587.8 kb)
|EU Declaration of Conformity|
EE354 CE-DoC(60.4 kb)
Declaration of Conformity to EU REACH(366.08 kb)
Material: sintered stainless
Pores size: 10 µm
Length: 33mm (1.30")
For high mechanical stress and strong pollution
T range: -40...180°C (-40...176°F)
Response time t10/90: 30s
Industrial process control
Life stock barns
Unsuitable for condensing environment
The main difference is the hardware interface. The Modbus RTU protocol runs on RS485 Hardware while Modbus TCP/IP protocol on Ethernet hardware.
The dew point temperature (Td) indicates where condensation begins. The frost point temperature (Td) indicates where icing begins.
Yes, the probe and the sensing element of all E+E dew point measurement devices is wettable.
The standard scope of supply of E+E dew point sensors includes an inspection certificate according DIN EN 10204 - 3.1. For details on calibration certificates please see page 5 in www.kalibrierdienst.at/fileadmin/content/White_Paper_Calibration_and_traceability_in_measuring_technology.pdf
Accredited calibration certificate for dew point temperature is available upon request, please see our calibration website.
Transmitter with "two-wires" technology receive the power from the process, and the signal is carry on return wire (closed current loop). With technology "three-wires" the power supply is separate from current output: 2 wires are the power supply and third carry the signal.