EE23은 Wall/Duct 또는 프로브 타입으로 다양하게 구성되어 있으며 설치현장에 적합한 타입으로 선택해서 사용할 수 있습니다. 모듈 형태의 외형으로 설치/교체 및 유지/관리가 간편하도록 디자인 되었습니다.
측정값은 아날로그, 알람 출력 및 LCD 디스플레이(액정)로 확인 할 수 있습니다. E+E의 독점기술인 센서코팅으로 외부의 오염 및 부식의 위험으로부터 측정 부를 안전하게 보호합니다.
EE23은 제품겉면의 설정버튼 또는 소프트웨어를 통해 설정 값 조정이 가능합니다.
-40...180 °C (-40...356°F)
±0.2 °C (±0.36 °F)
0-5/10 V, 0/4-20mAOutputs
24 V AC/DCSupply
EE23 Datasheet(715.92 kb)
Scaling of the outputs(252.78 kb)
Humidity Calibration-Kit Datasheet(642.96 kb)
EE23 Manual(1375.53 kb)
EE23 Calibration manual(86.26 kb)
Calibration Kit - User Guide(537.45 kb)
Cleaning Instructions(891.89 kb)
|EU Declaration of Conformity|
EE23 CE-DoC(324.74 kb)
Declaration of Conformity to EU REACH(366.08 kb)
Filter: PTFE membrane
Pores size: 1 µm
Length: 34mm (1.34")
Very good protection against fine dust
T range: -40...80°C (-40...176°F)
Response time t10/90: 15s
How can I clean the filter cap of a humidity measuring device?
The E+E propritary Coating helps in dirty, dusty and corrosive environment. salts, off-shore applications). It improves the sensor’s long term stability in dusty, dirty or oily applications by preventing stray impedances caused by deposits on the active sensor surface and on its connecting leads.
Coating does not help again chemicals in gazeous form. Contact an E+E sales representative for details on the influence of certain chemicals onto the humidity sensing elements.
Response time of a humidity sensor protected by the E+E proprietary coating is slightly longer than the response time without coating. Please see the data sheet of the specific sensor.
Temperature gradients along the sensing probes lead to relevant measurement errors. Whenever possible, install the entire probe inside the space to be monitored. If the probe is installed into a separation wall, make sure to isolate thermally the part of the probe looking out of the wall at the cable side.
The product data sheet indicates if and what type of inspection certificate is included in to standard scope of supply. Most of the E+E measurement devices feature an inspection certificate according to DIN EN 10204-3.1 or DIN EN 10204-2.2. An accredited calibration certificate is available on request, see the E+E scope of accreditation at our calibration website.
The calibration / adjustement interval depend from the application and from the requirements of the specific industry. See also "Princibles of Humidity measurement".
Transmitter with "two-wires" technology receive the power from the process, and the signal is carry on return wire (closed current loop). With technology "three-wires" the power supply is separate from current output: 2 wires are the power supply and third carry the signal.