EE364 is a compact and rugged moisture in oil transmitter for inline oil monitoring. It is optimised for high accuracy and excellent long term stability in lubrication, hydraulic and insulation oils, as well as diesel fuel.
Water activity (aw) and temperature (t) are accurately measured and the water content (x) in ppm precisely calculated. EE364 features two 4-20 mA outputs and one Modbus RTU interface.
With an optional confugration cable it is possible to perform an adjustment, set the MODBUS interface parameters and configure the analogue outputs.
- Measurands: water activity (aw), temperature and water content (x) in ppm
- Oil temperature up to 100 °C (212 °F)
- Two 4...20 mA outputs
- Modbus RTU interface
- Pressure rating 20 bar
- G ½" ISO or ½" NPT process connection
- Outputs freely configurable by the user
- Monitoring of
- Transformer oil
- Lubrication oil
- Hydraulic oil
- Diesel fuel
0...1 awMeasuring range
-40...80 °C (-40...176 °F) or
-40...100 °C (-40...212 °F)
±0.2 °C (0.36 °F)
2 x 4-20 mA / MODBUS RTUOuputs
10-28 V DCSupply
Send us your e-mail, we will do our best to help you!
EE364 Datasheet(780.31 kb)
Scaling of the outputs(252.78 kb)
EE364 Manual(859.33 kb)
Cleaning Instructions(1054.21 kb)
Relabeling in case of product configuration change(1339.66 kb)
Modbus Application Note AN0103(158.85 kb)
EE-PCS Product Configuration Software(28193.2 kb)
Stainless steel filter
What is the difference between Modbus RTU and Modbus TCP/IP?
The main difference is the hardware interface. The Modbus RTU protocol runs on RS485 Hardware while Modbus TCP/IP protocol on Ethernet hardware.
What is the calibration for a specific oil? What are specific oil parameter?
Calibration for a specific oil is a laboratory procedure for determining the saturation curve of the oil. The saturation curved is described by two parameters (A and B). These parameters are used for setting up the moisture in oil sensor for correct calculation of the water content (ppm) out of the measured temperature and water activity (aw).
How can I replace the sensing element of the E+E MIO sensor?
Sensor replacement is possible the EE360 models when option PC4 (pluggable probe). Please see the operation manual for detail and spare part code ordering guide.
How can I clean the sensing element of the E+E MIO sensor?
Please see the cleaning instructions.
Can the coating be used for E+E MIO sensors?
No, coating is not available for MIO sensors.
What kind of calibration certificate is included in the standard scope of supply of the E+E MIO sensors?
The standard scope of supply of E+E MIO sensorsincludes an inspection certificate according DIN EN 10204 - 3.1. For details on calibration certificates please see our white paper "Calibration and Traceability in Measuring Technology".
Are there different A, B parameters for mineral transformer oil from various manufacturers?
No, mineral transformer oils from different manufacturers are quite similar.. A and B parameters are the same.
What is the difference between a 2 wire and a 3 wire current output?