The EE772 is a thermal mass flow meter for compressed air and gases. Irrespective of pressure and temperature, the EE772 flow meter registers mass flow, volumetric flow or standard flow.
The flow meter can be used effectively to measure the consumption of compressed air, nitrogen, CO2 or other non-corrosive gases.
The gauge mounting block allows for the installation and removal without flow interruption. Regular calibration, without taking into account the device downtime, is therefore always an option. Besides the mass flow probe, the gauge mounting block with hot tap valve can accomodate also a dew point and a pressure sensor.
EE772 flow meter employs the well proven E+E hot-film sensor element with high immunity to contamination, which leads to outstanding long-term stability and accuracy even under harsh operating conditions.
With wide measuring range from 0.5 to 200 Nm/s (100...39370 SFPM) (1:400) and fast response time, one and the same flow meter model can be employed throughout the entire gas supply system. It records accurately even the smallest flow which represent an essential prerequisite for calculating consumption costs or for finding leaks.
Every E+E flow meter undergoes a precise factory calibration procedure at a pressure of 7 bar (102 psi), the corresponding calibration certificate is included in the standard scope of supply. The EE772 flow meter is available with an additional MODBUS RTU or M-BUS (Meter-Bus) interface.
- Gauge mounting block with hot tap valve for rapid installation and removal without flow interruption
- Additional installation of dew point- and pressure sensor
- Remote, interchangeable probe for easy mounting
- Pipe diameter DN40 - DN80 (1 1/2" - 3")
- Pressure range up to 40 bar (580 psi) (PN40)
- wide measuring range up to 200 Nm/s (39370 SFPM)
- MODBUS RTU or M-BUS
standardized volumetric flow, mass flow, standardised flow, temperature, consumptionMeasured values
0.5 - 200 Nm/s (100…39370 SFPM)Measuring range
±1.5 % of the measured valueAccuracy
DN40 up to DN80 (1 1/2" - 3")Pipe diameter
0-10 V, 0/4-20 mA, Pulse, Switch, Modbus RTU or M-BusOutputs
18 - 30 V AC/DCSupply
Send us your e-mail, we will do our best to help you!
EE772 Datasheet(1516.14 kb)
EE772 Manual(6134.77 kb)
Relabeling in case of product configuration change(1339.66 kb)
EE77x Configuration Software(8710.28 kb)
Modbus Application Note AN0103(158.85 kb)
Modular, compact in-line flow meter for DN15 to DN50
The modular inline flow meter is dedicated for metering and monitoring of compressed air and technical gases in DN15 to DN50 (1/2" - 2") pipes. Outstanding measuring accuracy due to multi-point factory adjustment at 7 bar (102psi) air pressure.
Inline mass flow meter DN15 - DN50
The inline mass flow meter for DN15 - DN50 (1/2" - 2") and up to 16 bar (232 psi). The measurement ball valve permits rapid installation and removal with only a short interruption of the flow.
Insertion mass flow meter for DN50 - DN700 (2" - 28") and up to 16 bar (232 psi)
Insertion mass flow meter with patented non-return protection for secure mounting. Use in pipelines DN50 up to DN700 (2" to 28"). Installation and removal without interruption of the flow.
Connecting Cable, 5 pins - 2m
Connecting Cable, 5 pins - 5 m
Connecting Cable, 5 pins - 10 m
What is the difference between Modbus RTU and Modbus TCP/IP?
The main difference is the hardware interface. The Modbus RTU protocol runs on RS485 Hardware while Modbus TCP/IP protocol on Ethernet hardware.
What calibration certificate is included in the standard scope of supply of the E+E mass flow sensors?
The standard scope of supply of E+E mass flow measurement devices includes an inspection certificate according DIN EN 10204 - 3.1. For details on calibration certificates please see our white paper "Calibration and Traceability in Measuring Technology".
Can E+E supply an accredited calibration certificate with the mass flow transmitter?
Accredited calibration certificate formass flow is available upon request, please see our calibration website.
How can I adjust the mass flow transmitter?
For readjusting the mass flow sensor you need first of all a reliable reference. In case this is not available, please return the device to E+E for service. Otherwise see the readjustment procedure in the operation manual of the specific E+E mass flow sensor available at www.epluse.com
E+E mass flow measure the Normal Volume referring normal conditions 0 °C and 1.013 mbar (14,7 psi, 32 °F), according standard DIN 1343. Can I change the normalization parameters according my standard definitions?
Yes. Use the E+E Product Configuration Software, section "Process Parameters". See operation manual of the specific E+E device.
What is the difference between a 2 wire and a 3 wire current output?
Transmitter with "two-wires" technology receive the power from the process, and the signal is carry on return wire (closed current loop). With technology "three-wires" the power supply is separate from current output: 2 wires are the power supply and third carry the signal.