The EE75 air velocity transmitters are optimized for high accuracy up to 40 m/s (8000 ft/min) over the temperature range -40...120 °C (-40...248 °F).
The high-end air velocity sensor element operates on the hot film anemometer principle and which stands out by high sensitivity even at very low air speed and high insensvity to pollution. T
he robust sensing probe and enclosure allow for EE75 use in harsh industrial environment as well as in applications with pressure rating up to 10 bar (145 psi).
Beside measuring air velocity and temperature, EE75 calculates the volumetric flow rate in m³/h or ft³/min based on the cross section of the duct, with an appropriate factory correction. The EE75 can be used to measure the velocity of various non-flammable and non-corrosive gases.
0...2 m/s (0...400 ft/min)Measuring range Air Velocity
0...10 m/s (0...2000 ft/min)
0...40 m/s (0...8000 ft/min)
0.06...2 m/s: ±0.03 m/sAccuracy Air Velocity
0.15...10 m/s: ±(0.1 m/s + 1% of mv)
0.2...40 m/s: ±(0.2 m/s + 1% of mv)
-40...120 °C (-40...248°F)Measuring range Temperature
±0.5 °C (±0.9 °F)Accuracy Temperature
0...40 m/s (0...8000 ft/min)Operating range
-40...120 °C (-40...248 °F)
pressure-tight models up to max. 10bar
freely scaleable in the ranges ofOutputs
0-10V and 0-20mA (e.g. 4...20mA)
Send us your e-mail, we will do our best to help you!
EE75 Datasheet(1268.49 kb)
Scaling of the outputs(252.78 kb)
EE75 Manual(2003.84 kb)
Relabeling in case of product configuration change(1339.66 kb)
EE7x Configuration Software(10513.37 kb)
EE75 Calibration manual(231.47 kb)
EE75 Manual USB connection(39.24 kb)
How can I clean the air velocity sensing element?
Due to its design and materials, the E+E thin film sensing element for air velocity is highly insensitive to pollution. When operaten in very dusty or diry environment, cleaning might be necessary. See section maintenance at downloads.epluse.com/fileadmin/data/miscellaneous/datasheet_air_velocity_measurement_basics.pdf
How can I replace the air velocity sensing element or the air velocity sensing probe?
This is not possible. Return the device to E+E for service.
What kind of calibration certificate is included in the standard scope of supply of the E+E air velocity sensors?
The standard scope of supply of E+E air velocity sensors includes an inspection certificate according DIN EN 10204 - 2.2 or DIN EN 10204 - 3.1, depending on the model. Please see the product data sheet and operation manual for the type of certificate. For details on calibration certificates please see page 5 in www.kalibrierdienst.at/fileadmin/content/White_Paper_Calibration_and_traceability_in_measuring_technology.pdf
Can E+E supply an accredited calibration certificate with the air velocity sensor?
Accredited calibration certificate for air velocity is available upon request, please see our calibration website.
How I can readjust my E+E air velocity sensor?
For readjusting the air velocity sensor you need first of all a reliable reference. This would be a wind channel. For the readjustment procedure please see the operation manual of the specific E+E air velocity sensor available at www.epluse.com
In case an approriate air velocity reference in not available, please return the device to E+E for service.
What is the difference between a 2 wire and a 3 wire current output?
Transmitter with "two-wires" technology receive the power from the process, and the signal is carry on return wire (closed current loop). With technology "three-wires" the power supply is separate from current output: 2 wires are the power supply and third carry the signal.