The EE060 and EE061 are compact humidity and temperature probes with excellent price/performance ratio, optimized for OEM applications.
EE060 features voltage outputs (0-1/5/10 V) for humidity and temperature, as well as optional passive T output.
EE061 has 4-20 mA output for humidity and a passive T output.
With encapsulated measurement electronics inside the sensing probe and E+E proprietary coating protection of the sensors, the EE060 and EE061 probes offer outstanding long term stability and reliability even in heavily polluted applications. With an optional radiation shield the probes can be also employed outdoors.
0...100% RHMeasuring range
0-1/5/10 V or 4-20 mAOutput humidity
0-1/5/10 V and passive T outputOutput temperature
EE060: 3.6-30 V DCSupply
EE061: 9-28 V DC
Send us your e-mail, we will do our best to help you!
EE060 Datasheet(887.24 kb)
EE061 Datasheet(512.08 kb)
Humidity Calibration-Kit Datasheet(428.89 kb)
EE060 Manual(581.25 kb)
EE061 Manual(265.61 kb)
Calibration Kit - User Guide(1042.14 kb)
Cleaning Instructions(1054.21 kb)
Passive T sensors
R-T Characteristics NTC10k - B25/85:3989K(265.4 kb)
Humidity and Temperature Probe with Digital Output
Compact design, digital output and M12x1 connector facilitate easy installation and replacement of the EE07 humidity and temperature probe.
High Precision Humidity and Temperature Probe
High measurement accuracy over a wide working range and low power consumption for indoor and outdoor measurements.
Filter: PTFE membrane
Pores size: 1 µm
Length: 34mm (1.34")
Very good protection against fine dust
T range: -40...80°C (-40...176°F)
Response time t10/90: 15s
Suitable for Products:
Flange Coupling, 5 pins
Connecting Cable, 5 pins - 2m
Connecting Cable, 5 pins - 5 m
Connecting Cable, 5 pins - 10 m
Connecting Cable, 5 pins, flying leads - 1.5m
Connecting Cable, 5 pins, flying leads - 5m
Connecting Cable, 5 pins, flying leads - 10m
What is the difference between a 2 wire and a 3 wire current output?
Transmitter with "two-wires" technology receive the power from the process, and the signal is carry on return wire (closed current loop). With technology "three-wires" the power supply is separate from current output: 2 wires are the power supply and third carry the signal.